Year : 2017 | Volume
: 10 | Issue : 2 | Page : 131--136
Patent ductus arteriosus closure using Occlutech® Duct Occluder, experience in Port Elizabeth, South Africa
Lungile Pepeta, Adele Greyling, Mahlubandile Fintan Nxele, Zongezile Masonwabe Makrexeni
Department of Paediatrics and Child Health, Division of Paediatric Cardiology, Dora Nginza Hospital, Walter Sisulu University, Port Elizabeth, South Africa
Background: Percutaneous closure of patent ductus arteriosus (PDA) has become standard therapy. Experience with the Occlutech® Duct Occluder is limited.
Methods: Data regarding ductal closure using Occlutech® Duct Occluder were reviewed and prospectively collected. Demographics, hemodynamic and angiographic characteristics, complications, and outcomes were documented.
Results: From March 2013 to June 2016, 65 patients (43 females and 22 males) underwent percutaneous closure of the PDA using Occlutech® Duct Occluder. The median age of the patients was 11 months (range, 1–454 months) and the median weight was 8.5 kg (range 2.5–78 kg). The mean pulmonary artery median pressure was 27 mmHg (range, 12–100 mmHg) and the QP: Qs ratio median was 1.8 (range, 1–7.5), with a pulmonary vascular resistance mean of 2.7 WU (standard deviation [SD] ±2.1). Thirty-two patients had Krichenko Type A duct (49%); 7, Type C (11%); 4, Type D (6%); and 22, Type E (34%). The ductal size (narrowest diameter at the pulmonic end) mean was 3.5 mm (SD ± 1.9 mm). The screening time mean was 17.3 min (SD ± 11.6). Out of 63 patients with successful closure of the PDA using Occlutech® Duct Occluder, there were 15 patients with small PDAs; 25 with moderate PDAs, and 23 with large PDAs. In one patient, the device dislodged to the descending aorta, and in two patients, to the right pulmonary artery immediately following deployment, with successful percutaneous (two) and surgical (one) retrieval. Complete ductal occlusion was achieved in all 63 patients on day one.
Conclusion: The Occlutech® Duct Occluder is a safe and effective device for closure of ducts in appropriately selected patients.
Faculty of Health Sciences, Building 7, 1st Floor, R0126, Summerstrand Campus (South), Nelson Mandela Metropolitan University, Port Elizabeth 6001
|How to cite this article:|
Pepeta L, Greyling A, Nxele MF, Makrexeni ZM. Patent ductus arteriosus closure using Occlutech® Duct Occluder, experience in Port Elizabeth, South Africa.Ann Pediatr Card 2017;10:131-136
|How to cite this URL:|
Pepeta L, Greyling A, Nxele MF, Makrexeni ZM. Patent ductus arteriosus closure using Occlutech® Duct Occluder, experience in Port Elizabeth, South Africa. Ann Pediatr Card [serial online] 2017 [cited 2021 May 15 ];10:131-136
Available from: https://www.annalspc.com/article.asp?issn=0974-2069;year=2017;volume=10;issue=2;spage=131;epage=136;aulast=Pepeta;type=0